How to clean, sterilize and sterilize the Veress Needle？
After the product is used, it should be cleaned immediately with tap water. The dismantled part should be disassembled as far as possible, and all parts of the apparatus should be washed under tap water to thoroughly clean the dirt or bloodstain left after the operation, and then dry it with clean gauze. Before disinfecting the products, use tap water to rinse thoroughly and dry the instruments with clean gauze.
Disinfection can be used in high temperature, high pressure, or soaking and disinfection. A) high temperature and high pressure disinfection: 134 centigrade, 2.0bar. B) soak disinfection: immerse all the instruments in a solution of 2% alkaline glutaraldehyde for 45 minutes or longer.
The sterilized instruments are dry with 75% ethanol or clean compressed air.
Sterilization can be sterilized with pressure steam, ethylene oxide sterilization, and 2% alkaline glutaraldehyde solution for 10 hours sterilization.
ZHEJIANG SHENDASIAO MEDICAL INSTRUMENT CO.,LTD
Add：NO.699, EAST BAIYUNYUAN ROAD, TONGLU, HANGZHOU, ZHEJIANG,CHINA